Micro Mega - One Shape Procedure Files Pack
Warning: Last items in stock!
Buying this product you will collect 0 points with our loyalty program.
Information of seller
One Shape Procedure
Endo Flare: #25 - T=12% - L=15mm
MMC (K File): #10 - L=25mm
One G: #14 T=3% L=25mm
MMC (K File): #15 - L=25mm
One Shape: #25 - T=6% - L=25mm
One Shape is a single NiTi file for root canal shaping. It is effective for canals with high curve and its safe tip helps easy entry to the apical and also prevents the closing of the canal which is a fracture factor.
One Shape has 3 different cross- section zones:
The first zone presents a variable 3-cutting-edge design. The second (transition zone) has a cross-section that progressively changes from 3 to 2 cutting edges. The last (coronal) is provided with 2 cutting edges.
One Shape advantages:
- Simple use ---> One System File
- Time saving ---> Approximately 60% faster than a conventional treatment
- Flexibility for the practitioner ---> Great ability to negotiate curves.
- Safety ---> Single use and being sterile.
- Controls risk of infection due to sterile blister packaging
- Innovation ---> Asymmetrical cross-section with longer pitch
- Ergonomic ---> Maximum comfort, safety and Efficiency
Operating protocol of One Shape:
1. Access cavity preparation:
- Take a preoperative radiograph to assess canal anatomy and establish Estimated Working Length (EWL)
- Establish straight-line access preparation to expose canal orifices.
- Remove coronal constrictions with ENDOFLARE (or use your current flaring technique). Canal penetration of ENDOFLARE is limited to 3 mm below the pulp chamber floor.
2. Establishing the glide path:
Scout each root canal with small diameter hand files which provide information about the root canal anatomy complementary to that obtained by the preoperative X-rays. Prepare the glide path using #10 Stainless Steel K-type hand file (for example MMC 10)
**If the #10 K file easily reaches the EWL, and if it is possible to take a #15 K file down to this EWL:
If canal constrictions prevent the #10 K file from easily reaching the EWL: complete glide path preparation using G-Files™, NiTi files for glide path development in continuous rotation. Use G1 to EWL followed by G2 to EWL:
Use the #15 K file to determine Working Length (WL).
3. Shaping the root canal:
- Irrigate thoroughly with sodium hypochlorite. A chelating gel (MM-EDTA Cream) can also be used during the root canal shaping.
- Rotation speed: 350 – 450 tr/min – Max torque: 2.5Ncm
- Technique: in and out movement without pressure.
- Remove and clean the instrument and irrigate the canal when apical resistance or a slight apical pull is encountered.
- This instrument is for single use only and has been designed for the treatment of only one tooth.
- Check the condition of the instrument after the treatment of each canal and dispose of all instruments which show signs of unwinding, wear or premature fatigue. Multiple usage will increase the risk of separation.
|File Number||Endo Flare 25/.12 | K File #10 | K File #15 | G File | One Shape File|
|Number Of Files In Pakage||5|
|Taper||0.12 | 0.03 | 0.06|
|Length||15 | 25|
|Torque range||Max. 2.5Ncm For One Shape|
|Type||Single File System|
|Speed of Rotation||350-450rpm for One Shape|
No customer reviews for the moment.
NO registration required!
If the question you have has not yet been answered here, use the form below to ask something about this addon.